1). Output frequency
Frequency range: 0.2Hz~2MHz seven ranges:0.2Hz-2Hz, 2Hz-20Hz, 20Hz-200Hz, 200Hz-2kHz, 2kHz-20kHz, 20kHz-200kHz, 200kHz-2MHz,
Output signal impedance: 50Ω
Output signal wave-form: sine wave, square wave, rectangle wave, sawtooth wave and triangle wave
Output signal amplitude (peak-peak value): non-attenuate (2Vp-p~20Vp-p)±20% continuously adjustable
attenuate 20dB (0.2Vp-p~2.0Vp-p)±20% continuously adjustable
attenuate 40dB (20mVp-p~200mVp-p)±20% continuously adjustable
The above are measured with load 1MΩ, the output signal amplitude will be half of standard at 50Ω load.
Function output symmetry adjust scope: 20%~80% (±10%)
6). Output signal features: sine wave distortion:＜2%, triangle wave linear:＞99% (10%-90% of output amplitude), square wave rise edge times: less than 100nS (10%-90% of output amplitude), square wave fall edge times: less than 100nS (10%-90% of output amplitude), square wave rise and fall pulse less than or equal to 5%Vo (50Ωload).
Test condition: frequency output: 10KHz, amplitude: 5Vp-p,warm-up for 20minutes.
Output signal frequency stability: less than ±0.1%/min (test condition is the same as the above)
Amplitude display (only for 50Ω load, at 1MΩ load, the real output amplitude is double of the displaying value):
Display digits:2/3 digits (decimal point automatic select place). Display units: Vp-p or mVp-p. Display errors: Vo±10%±1d (Vo refers to the true value of output signal) Resolution: non-attenuate 0.2Vp-p 20dB attenuate:20mVp-p 40dB attenuate:2mVp-p
display range: 0.2Hz-2MHz
display effective digit: four or five digits.
Measurement errors: ≤0.5%
Time base: frequency: 12MHz
Working temperature: 0℃~40℃
Size: 270mm x 215mm x 100mm
Power applicability and consume: 110V/220V±10% 50Hz/60Hz±5%, power consume ≤15W
It's convenient to operate and use with high intelligence for using the single microprocessor to control running and displaying.
Large scale single integrated accuracy function generator leads to super performance.
Designed by large scale integrated circuit to insure the high reliability and high stability.
Preheat for at least 10 minutes before operation.