# Milliamp Meter to Voltmeter

Posted on Nov 25, 2012

A milliamp meter can be used as a volt meter by adding a series resistance. The resistance needed is the full scale voltage reading divided by the full scale current of the meter movement. So, if you have a 1 milliamp meter and you want to read 0-10 volts you will need a total resistance of 10/.001 = 10K ohms. The meter movement itself will have a small resistance which will be part of the total 10K resistance, but it is usually low enough to ignore. The meter in the example below has a resistance of 86 ohms so the true resistor value needed would be 10K-86 or 9914 ohms.

But using a 10K standard value will be within 1% so we can ignore the 86 ohms. For a full scale reading of 1 volt, the meter resistnace would be more significant since it would be about 8% of the total 1K needed, so you would probably want to use a 914 ohm resistor, or 910 standard value. The milliamp meter can also be used to measure higher currents by adding a parallel resistance. The meter resistance now becomes very significant since to increase the range by a factor of ten, we need to bypass 9/10 of the total current with the parallel resistor. So, to convert the 1 milliamp meter to a 10 milliamp meter, we will need a parallel resistor of 86/9 = 9.56 ohms.

Leave Comment characters left:

• ## New Circuits

.

Popular Circuits

Square-wave-pulse-extractor
pcb How to use the same switch to control LED and vary resistance
thrifty voltage regulator
astable multivibrator
vcd
555 organ circuit
remote control sckt
Strobe Flash Light Circuits
LM1881 LM1881 Video Sync Separator
Touch Sensor Circuit
Circuit Enables Single NiMH Cell to Simulate a Lithium Battery
TEMPERATURE REPORTING DIGITAL THERMOMETER
lm358 alkaline battery charger circuit

Top 